Modest of Operations: AED mode, automatic ECG analysis with audio and visual instructions, manual mode, asynchronous and synchronous defibrillation per paddles or self-adhesive electrodes, ECG monitoring and pacing.
TRANSTHORACIC PACEMAKER (NIP)
Impulse Shape: Biphasic
Energy in the automatic mode: According to patient impedence
Energy levels in manual mode: 1J, 2J, 3J,4J, 5J, 6J, 7J, 8J, 9J, 10J, 15J, 20J, 30J, 40J, 50J, 75J, 100J, 125J, 150J, 175J, 200J, 300J, 360J,
Charge time: < 6s for 200 J (with a new fully charged battery at 25◦c) 8s for 360 J (with new fully charged battery at 25 ◦c)
Pacemaker Detection: Detect pacer pulses of ±2 mV to ±700 mV with pulse widths of 0.1 msec to 2 msec and a rise time of 10 % of the width but not more than 100 μs
Cardioversion: Manual Activation of the cardioversion via the SYNC button, defibrillation via paddles as well as self-adhesive electrodes.
Model: TFT-LCD color, dual mode
Dimensions: 170 x 128 mm (diagonal 8.4“/ 212 mm)
Resolutions: 800 x 600 pixel
Number of channels: 4 waveforms or 12 ECG channels, cascaded can be shown
Pulse oximeter (SpO2)*
Range: 1 % bis 100 %
Pulse Rate: 20 – 300 bpm
Accuracy: 70% – 100%“
Operation mode: Manual and automatic measurement of interval
Technology: Oscillometric measurement
Measurement range neonatal: SYS 40 to 120 mmHg / MAP 30 to 100 mmHg / DIA 20 to 90 mmHg
Measurement range: -50 – 300 mmHg
Channels: 2 parallel connections
Type/EtCO2 –Module: Respironics® / Mainstream / Sidestream
Ambient-CO2-level: 0 – 150 mmHg
Parameter: EtCO2, InCO2
You can contact us
Your message has been successfully sent. We will contact you very soon!
There was a problem with your form submission.
Please try again and make sure all required fields are filled out correctly.